This is for the first time that application of complex nanostructure is reported as pH indicator in PVC matrix. This new optical pH sensor was constructed based on incorporation of ZnLI2 complex nanostructure in PVC matrix. The synthesized nanostructure ZnLI2 complex was characterized by SEM and XRD technique. The membrane solution was speared on the glass plate to provide thin film and the membrane surface morphology was investigated via field emission scanning microscope (FE‐SEM) technique. Central composite design (CCD) combined with desirability function (DF) was applied to find the best experimental composition of membrane providing the highest absorbance. These conditions were found in correspondence with 3 mg of pH indicator, 3 mg of ionic additive and 1.5 mg/mg of DBP/PVC weight ratio. Under optimum conditions, the proposed pH sensor has two linear working ranges of 4 ‐ 8 at 393 nm (R2 = 0.9897) and 5 ‐ 8 (R2 = 0.9982) at 570 nm with response time of 4 min. The pKa of proposed pH optical sensor was calculated through three methods that found to be 5.63. The present optical sensor shows stability after 2 months without any significant divergence in response properties (less than 5% RSD). Furthermore, current pH optode was exhibited good repeatability (RSD = 1.14%) as well as reproducibility (RSD = 4.06%). No significant variation was observed on sensor response with increasing the ionic strength in the range of 0.0–0.5 M of sodium chloride. All above features indicated that the proposed sensor can be successfully used for detection of pH in solutions with different ionic strength.
Applied Organometallic Chemistry – Wiley
Published: Jan 1, 2018
It’s your single place to instantly
discover and read the research
that matters to you.
Enjoy affordable access to
over 12 million articles from more than
10,000 peer-reviewed journals.
All for just $49/month
It’s easy to organize your research with our built-in tools.
All the latest content is available, no embargo periods.
“Whoa! It’s like Spotify but for academic articles.”@Phil_Robichaud