# Research on Intelligent Control System of DC SQUID Magnetometer Parameters for Multi-channel System

Research on Intelligent Control System of DC SQUID Magnetometer Parameters for Multi-channel System In a multi-channel SQUID measurement system, adjusting device parameters to optimal condition for all channels is time-consuming. In this paper, an intelligent control system is presented to determine the optimal working point of devices which is automatic and more efficient comparing to the manual one. An optimal working point searching algorithm is introduced as the core component of the control system. In this algorithm, the bias voltage $$V_\mathrm{bias}$$ V bias is step scanned to obtain the maximal value of the peak-to-peak current value $$I_\mathrm{pp}$$ I pp of the SQUID magnetometer modulation curve. We choose this point as the optimal one. Using the above control system, more than 30 weakly damped SQUID magnetometers with area of $$5 \times 5 \, \hbox {mm}^2$$ 5 × 5 mm 2 or $$10 \times 10 \, \hbox {mm}^2$$ 10 × 10 mm 2 are adjusted and a 36-channel magnetocardiography system perfectly worked in a magnetically shielded room. The average white flux noise is $$15 \, {\upmu \Phi }_0/\hbox {Hz}^{1/2}$$ 15 μ Φ 0 / Hz 1 / 2 . http://www.deepdyve.com/assets/images/DeepDyve-Logo-lg.png Journal of Low Temperature Physics Springer Journals

# Research on Intelligent Control System of DC SQUID Magnetometer Parameters for Multi-channel System

8 pages

/lp/springer_journal/research-on-intelligent-control-system-of-dc-squid-magnetometer-LLj7KcFN1N
Publisher
Springer Journals
Subject
Physics; Condensed Matter Physics; Characterization and Evaluation of Materials; Magnetism, Magnetic Materials
ISSN
0022-2291
eISSN
1573-7357
DOI
10.1007/s10909-018-1881-8
Publisher site
See Article on Publisher Site

### Abstract

In a multi-channel SQUID measurement system, adjusting device parameters to optimal condition for all channels is time-consuming. In this paper, an intelligent control system is presented to determine the optimal working point of devices which is automatic and more efficient comparing to the manual one. An optimal working point searching algorithm is introduced as the core component of the control system. In this algorithm, the bias voltage $$V_\mathrm{bias}$$ V bias is step scanned to obtain the maximal value of the peak-to-peak current value $$I_\mathrm{pp}$$ I pp of the SQUID magnetometer modulation curve. We choose this point as the optimal one. Using the above control system, more than 30 weakly damped SQUID magnetometers with area of $$5 \times 5 \, \hbox {mm}^2$$ 5 × 5 mm 2 or $$10 \times 10 \, \hbox {mm}^2$$ 10 × 10 mm 2 are adjusted and a 36-channel magnetocardiography system perfectly worked in a magnetically shielded room. The average white flux noise is $$15 \, {\upmu \Phi }_0/\hbox {Hz}^{1/2}$$ 15 μ Φ 0 / Hz 1 / 2 .

### Journal

Journal of Low Temperature PhysicsSpringer Journals

Published: Mar 5, 2018

## You’re reading a free preview. Subscribe to read the entire article.

### DeepDyve is your personal research library

It’s your single place to instantly
that matters to you.

over 18 million articles from more than
15,000 peer-reviewed journals.

All for just $49/month ### Explore the DeepDyve Library ### Search Query the DeepDyve database, plus search all of PubMed and Google Scholar seamlessly ### Organize Save any article or search result from DeepDyve, PubMed, and Google Scholar... all in one place. ### Access Get unlimited, online access to over 18 million full-text articles from more than 15,000 scientific journals. ### Your journals are on DeepDyve Read from thousands of the leading scholarly journals from SpringerNature, Wiley-Blackwell, Oxford University Press and more. All the latest content is available, no embargo periods. DeepDyve ### Freelancer DeepDyve ### Pro Price FREE$49/month
\$360/year

Save searches from
PubMed

Create folders to

Export folders, citations

Abstract access only

18 million full-text articles

Print

20 pages / month

PDF Discount

20% off